DP-560B5 Economical hospital Anesthesia machineAnesthesia Machine, with its ergonomic design, integrates a concise and smooth structure and high mobility. The microcomputer-controlled Ventilator adopts exported sensors and other components. Its scientific and proper design as well as rigorous production process are what demonstrate the inner quality of Aokai products: mechanical ventilation is made secure and visible by the folded respiratory bellows, and the security of air supply is guaranteed by pneumatic transmission and measurement system. With multiple modern technologies applied, the Anesthesia Machine is both outstanding in performance and versatile in functions.
(1)The Ventilator adopts microcomputer control with high-definition 10.4" TFT color LCD display, integrating the functions of respiratory control and displaying the monitoring parameters. It is versatile in function and excellent in performance.
(2)The high-precision Vaporizer is designed with a stable concentration output, automatic compensation for pressure, temperature and flow rate. And it is available to choose two kinds of gas from Halothane, Enflurane, Isoflurane, and Sevoflurane, ensuring a successful anesthesia operation.
(3)High-definition five-tube flowmeter (oxygen, nitrous oxide, and air), designed with stoppers for oxygen and nitrous oxide to control the output oxygen concentration.
(4)Reliable pneumatic transmission system.
(5)Economic and proper configuration of technologies, optional as required by the Customer.
(6)Modularized design and upgrading. Upgrading is available as required by the Customer.
|Part of Anesthesia Machine|
|Main body ||High-strength engineering plastic rack, light, |
beautiful and corrosion resistant
|Scope of application || adult, child|
|Gas source||O2, N2O, Air: 0.28~0.6MPa|
|Flow meter||O2 : 0.05~1.0L/min 1.1~10L/min|
N20: 0. 05~1.0L/min 1.1~10L/min
Air: 0. 05~10L/min
|O2, N2O linkage and N2O Stopper||When using nitrous oxide, oxygen concentration>25%;|
When the oxygen pressure<0.2Kpa, the flow of nitrous oxide would be cut off.
|Flow rate of rapid oxygen supply||25~75L/min|
|Low oxygen pressure alarm ||There will be sound alarm when the oxygen pressure < 0.2MPa|